Questions & Answers
Only trained clinicians should use the Ipas Manual Vacuum Aspirator. Please contact us to find out more about training and credentialing opportunities.
The total procedure can be done in 10 steps. Please see this chart for pictographic steps: https://ipas.azureedge.net/files/PERFMVAE17-PerformingMVAPoster.pdf
DKT WomanCare sells single-use variants of our products to comply with national regulations. Please ask your distributor whether your product is single-use or not.
Color-coded adapters allow Flexible Karman Cannula and 3 mm Cannula to be used with the Ipas MVA Plus(R) or Double Valve Aspirators. Please refer to the product technical sheet for more information.
Please refer to our Technical Resources under “Clean and Disassemble Instruments” or see this sheet from Ipas: https://ipas.azureedge.net/files/PROPLUSE18-ProcessingIpasMVAPlusAspiratorsEasyGripCannulae.pdf
Check that instrument is properly assembled and charged
Inspect O-ring for proper positioning - if damaged or loose, replace O-ring.
Too much lubrication.
Ensure no foreign bodies are present.
Check cylinder is firmly seated on valve.
Charge and test again.
If vacuum is still not retained, use another aspirator.
Check if :
Aspirator is full.
Cannula is withdrawn past the cervical os opening.
Cannula is clogged.
Aspirator is incorrectly assembled.
After the 25th use
The cylinder is cracked or brittle.
Mineral deposits inhibit plunger movement.
The Valve is cracked, bent or broken.
Buttons are broken.
Plunger arms do not lock.
Aspirator no longer holds a vacuum.